Background During the COVID-19 pandemic it is anticipated that there will be a shortage of mechanical ventilators available for patients in critical condition. This has sparked many discussions about rationing resources and withholding care; however, an alternative may be to implement manual ventilation in these situations instead. Manual ventilation and a safety device were assessed for efficacy of extended use, such as may be required during this pandemic.
Methods To evaluate physical output characteristics of extended manual ventilation and efficacy of a barotrauma mitigation device, 47 medical students, nurses and medics completed two 1-hour manual ventilation sessions using the SmartLung 2000 Lung Simulator and 5300 Series Mass Flow Meter with a SPUR II resuscitator bag and endotracheal tube, mimicking a healthy adult with normal lung physiology, both with and without the Sotair device. Providers were randomised to complete their initial session either with or without the Sotair device.
Findings Collected data show wide variability in tidal volume and peak pressure in unmitigated manual breaths despite prior training and independent exploration of the resuscitation equipment prior to testing. The mean (±SD) tidal volume with bag only was 563.9±128.8 mL and with the safety device 536.1±80.9 mL (p<0.0001). The mean peak inspiratory pressure with bag only was 17.2±6.3 cm H2O and with the safety device 14.9±2.4 cm H2O (p<0.0001).
Interpretation While extended manual ventilation cannot replace mechanical ventilation, it is feasible with a safety device, which may reduce barotrauma, underventilation and overventilation. These results also demonstrate that withholding care and rationing resources may not be necessary.
- critical care
- emergency medicine
- biomedical engineering
- pulmonary medicine
Data availability statement
All data relevant to the study are included in the article. Deidentified participant data are available upon request. Please contact Dr Mark Brady for permissions for reuse.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.