Introduction Bleeding during surgery is common. Increased bleeding may disturb procedure, induces haemodynamic instability and results in need for blood transfusion. Allogenic blood transfusions increase mortality and morbidity, especially risk of infections, pulmonary and renal complications, as well as thromboembolic events. Autotransfusion is in many cases a solution but forced suction may destroy or alter blood cells because of turbulences, shear forces and contact of the blood to extrinsic surfaces. The aim of the study was the analysis of turbulence profile and development of a new suction device reducing (or avoiding) turbulences.
Methods We registered turbulences with a microphone placed in different positions within the blood suction during surgery and analysed the spectrum. Then, we modified the circuit adding signals from optical sensors and pressure transducer to avoid air mixing and tight suction. Finally, we created the algorithm for the suction circuit regulating individualised suction modes.
Results We developed a new suction system based on a roller pump. We used a piezo sensor and registered the acoustic signals. The optimal position for this element was into the suction handle. After filtering the signal and further processing, we used it for regulation of the roller pump. Additionally, an optical sensor minimises air mixing due to further regulation of motor speed. Finally, a negative pressure transducer gives in case of tight suction information to the circuit stopping motor speed and equalising pressure by opening a valve before suctions starts again. The algorithm allows various suction modes in an individualised manner for specific situations in operating field.
Conclusions We developed a new blood suction device based on a roller pump. The system is turbulence-controlled and its algorithm allows several individualised suction modes. Additional features avoid tight suction and reduce air mixing.
- Blood Suction
- Blood Transfusion
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.